Is the website displaying in the correct language? Please confirm or select a different language.
The Basics of Load Cell Wiring and Trimming
Load cell wiring and trimming are critical to a weighing system’s accuracy and should always be performed by trained technicians. However, general load cell wiring knowledge can help you better understand how your weighing system works and how potential errors occur.
Load Cell Wiring Basics
Load cell cables typically have four or six wires. Both versions have positive and negative signal and excitation lines, though a six-wire cable also has positive and negative sense lines. Sense lines are linked to the sense connections of the system’s indicator, sometimes through a junction box, allowing the indicator to sense a load cell’s actual voltage.
If there is a voltage drop between the indicator and load cell, the sense lines send that information to the indicator, which automatically adjusts voltage. This compensates for the loss of voltage or amplifies the return signal to compensate for loss of power to the load cell.
Load cell wires are color coded to help ensure proper connections. Different brands use different color codes, which is why there are load cell wiring guides available for technicians to use during installations.
Understanding Load Cell Trimming
Load cell trimming is sometimes called load cell summing because it involves tying or summing multiple load cell output signals together. This happens in a junction box, also known as a summing box, which sends one system signal to the indicator.
It’s necessary to trim load cells in a multi-cell system when the weight distribution to each load cell isn’t equal. For example, in vessel loading processes, the presence of agitators or the type of material, such as powder, can affect how material and weight are distributed to each load cell. Additionally, it’s nearly impossible to make every load cell exactly alike. Tolerances in the manufacturing process allow for some variation in individual load cell specifications, which, if trimming isn’t applied, doesn’t allow for the accuracy necessary for most applications.
Types of Load Cell Trimming
There are two different types of load cell trimming that scale technicians rely on when installing multi-cell weighing systems.
Excitation trimming is the oldest method of trimming load cell output signals. Excitation trimming adds series resistance to the excitation circuit of the load cell, reducing excitation voltage at the load cell. The system load cell with the lowest output receives the full excitation voltage, while the other system load cells receive proportionately smaller excitation voltages. This creates matched outputs for all load cells in the system.
Signal trimming is the most common and popular type of trimming because it is the easiest to use. Signal trimming is compatible with nearly all indicators and is relatively unaffected by temperature changes or excessive system vibrations. It involves adding a relatively high parallel resistance between the signal of each load cell, which creates a leakage path that diverts some of the available load cell signal away from the indicator. More load cell signal will be available to the indicator as parallel resistance increases.
Rice Lake Weighing Systems’ Load Cell and Weigh Module Resource Center provides additional information on load cells and weigh modules, including links to resources such as Rice Lake’s Load Cell Wiring Guide and Load Cell and Weigh Module Handbook.
Advanced Load Cell Troubleshooting
Publish Date: 7/20/2017
Load cells are not only the most critical part of an electronic weighing system, they are also the most vulnerable.
What is a Load Cell and How Does it Work?
Publish Date: 12/29/2020
The load cell is the heart of every scale system. The load cell is the most important component of a digital scale system. All digital scales rely on load cells and without it, a scale can’t measure weight.
Operation, Capacity Versus Resolution, and Common Types of Load Cells
Publish Date: 7/15/2021
A load cell converts a mechanical force into an electrical signal. Learn how load cells work, the differences between load cell capacity and resolution, and some common types of load cells.
10 Facts About Load Cells
Publish Date: 9/1/2020
Load cells are truly the heart of every scale system. Learn 10 facts about load cells—from when load cell technology started all the way to the many different applications you can use them!
Choosing a Load Cell for Tough Applications
Publish Date: 5/4/2021
To select the right load cell for a demanding application, you need a solid understanding of your environment and operating conditions and what load cell features are best for handling them.
How Do I Know Which Load Cell I Need?
Publish Date: 8/29/2020
Choosing the right load cell for your application can be overwhelming. There are as many types of load cells as there are applications to use them. Here's a guide to help you through it.